Guide of Genex Probe for. LTE Drive Test 2 Huawei confidential, 2. Contents Overview of Genex Probe & Assistant Genex Probe. INTRODUCTION TO GENEX PROBE DRIVE TEST OPERATIONBY IMRAN HASHMI Huawei TECHNOLOGIES Co., Ltd. Huawei Confidential 1 GENEX. Huawei GENEX Probe & Assistant practice guide Internal use only Huawei GENEX Drive Test tools Practice Guide Abbreviation: User Manual: In Probe.
|Published (Last):||17 May 2015|
|PDF File Size:||10.80 Mb|
|ePub File Size:||5.91 Mb|
|Price:||Free* [*Free Regsitration Required]|
The legend color depends on the color of the parameter displayed in the Chart window. The screen pause function enables you to locate the problem. Set Item Property 1. Different items can be displayed based on the icons selected on the shortcut menu.
In such a situation, the frame number should be added with 1. You can perceive the change of the connection status in real time. For details about the import of area graphic information, refer to part 6. The string can be Message entered.
CH Code Genrx CH code refers to the channel code, which includes primary scrambling code, secondary scrambling code, and channel code, as shown in Figure After updating the license, connect the rest hard dongles. You can choose to clear the former stat.
The data of the same type is distinguished from each other with different colors. Measure Time z True: The UE link contains: The Probe can play forward or backward. Figure Failing to open the parameter pane z If a device to be displayed is of the same type as that selected in the Choose Device list, the Probe opens the parameter pane of the device.
Introduction to GENEX Probe – ppt download
The drive test route is generated by dotting the map during the indoor test. The time interval between multi downloads.
Can be set either Test by Count or Test by Time. To configure the hardware manually in the Probe and set the DTI Scanner parameters, perform the following steps: Table Description of tab operations To… You can… Add a tab Right-click the tab and choose Add on the shortcut menu.
Through geex of the following ways, you can trigger the window co-activation: If Modern port is not configed, Probe still can keep track of part of the data, but can not start PS domain test automatically by Testplan, can not keep track of the throughput on the application layer, either.
The Event dialog box is displayed, as shown in Figure For instance, when a device is unexpectedly Hardware alarm disconnected or the GPS signal is inadequate, the Probe warns you of the alarms in the form of either dialog box or sound.
If you select Parameter in the Layer Type area, you need to define map legends in the Parameter Layer dialog box. Specified The Specified sets the common properties of the data deive in the specified base station measurement.
The Choose Projection dialog box is Projection displayed, as shown in Figure Cable Table A-2 lists the cable parameters. The recommended value duration sec is 0. Means reader be extremely careful during the operation.
Introduction to GENEX Probe
This chapter describes the following: In this way, you can choose several drive test files recently used as your data source. Calibrate the propagation model by using the planning application software. The Probe records the prboe test data in the log file. The ManualConfig window is displayed.
In the former case, the options in the Size Type drop-down list are unavailable. You can control the test process and observe the data changes in the signaling window. Indicates the short message content. The co-activated information is gsnex if you click a record in a window like a signaling, map, or Co-activated information event window.