Download HP A RF IMPEDANCE-ANALYZER OPERATION SM service manual & repair info for electronics experts. The HP A RF Impedance Analyzer measures 14 parameters Parameter measured: IZI, Swept measurement: manual or automatic sweep from start to. Agilent A Test Fixture. Operation and Service Manual. Third Edition. Agilent Part No. February Printed in Japan.
|Published (Last):||3 April 2015|
|PDF File Size:||11.30 Mb|
|ePub File Size:||9.18 Mb|
|Price:||Free* [*Free Regsitration Required]|
Solved: GPIB Coding for the HPA Impedence Analyzer – Discussion Forums – National Instruments
Would you mind posting the code you used on the A? In the code above, MHz is an arbitrary frequency. Accepted by topic author Matt Hp4191q. Should I be looking at doing it with LabView or something else?
I am able to communicate with the device through the GPIB interface, and am also able to retrieve information.
Agilent / HP 4191A RF Impedance Analyzer
You really need the manual. Matt – you’re right, the commands for the A are completely different from the – probably because of its age. IIRC, the sweep parameters were something like this: Thanks for your help guys. Message 1 of 6.
Message 3 of 6. Message 5 of 6.
Message 4 of 6. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows: When I gave it a try, though, it worked fine. Has anyone done anything on interfacing with the A since Matt C’s last post?
A RF IMPEDANCE ANL [Obsolete] | Keysight (formerly Agilent’s Electronic Measurement)
I went through the manual before, but when I saw their GPIB section, the strings were so convoluted and so unlike anything I’ve ever seen before that I assumed it was referring to something different. Auto-suggest helps you quickly narrow down majual search results by suggesting possible matches as you type. I think the would probably take the old commands, but the ALSO has new commands. In particular, after setting up and initiating the instrument, I am initiating the measurement and reading it as follows:.
The forums emoticons took over for a bit there.
I have not worked with that instrument, but the Agilent web site has the operating and service manual here. Message 6 of 6.
Good luck with it Most Active Software Boards: Sorry, that last line of code there is supposed to read: Message 2 of 6. I suspect that this difference might be causing my problems. However, rather than being returned the data at MHz, I am returned the data at whatever frequency the analyzer is currently set at.
I have worked with the A and, as I recall, the commands are nothing like any other instrument I have seen. If not, what would creating a driver for it entail? Guess you never know until you try! It’s 84MB, so I hope you have a fast link.